In A026 says Magnus®-S3 chips come with single-point calibration
data pre-stored in the User memory bank. But AN02 talks about a two points in the user data structure. So which is it? because my RFM3254 only reads back 24 byts not 64.
Hi David, I am a S3 user, below is from my experience, hopefully can be useful to you.
The S3 is a two point calibration sensor, calibration data is storeed inside S3. what A026 means is they calibrate only 1 point in factory, the 2nd calibration data is using a default value, no further calibration in factory.